Alert number: A12/01004/20
Name: KN95 Stereo protective mask
Type of alert: Serious
Category: Protective equipment
Type / number of model: KN95
Risk type: Health risk / other
The product bears a CE marking but is not certified as protective equipment by a relevant body. Additionally, the particle/filter retention of the material is insufficient (measured value ≤ 86 %) and the mask does not properly adapt to the face leading to an insufficient total filtration capacity (measured value ≤ 80 %) . Consequently, an excessive amount of particles or microorganisms might pass through the mask, increasing the risk of infection if not combined with additional protective measures. / The product does not comply with the Personal Protective Equipment Regulation.
Measures ordered by public authorities (to: Other): Marking the product with appropriate warnings on the risks
Description: KN95 protective respiratory filtration half-face mask.
Packaging description: The product is sold in a white and blue plastic bag with zip closure containing 5 pieces.
Country of origin: People's Republic of China
Alert submitted by: Belgium