- Risk type
- Health risk / other
- Notifying country
- Alert number
The particle/filter retention of the material is insufficient (measured values: as low as to 41 %) Consequently, an excessive amount of particles or microorganisms might pass through the mask, increasing the risk of infection if not combined with additional protective measures.
The product does not comply with the Personal Protective Equipment Regulation and with the relevant European standard EN 149.
- Protective equipment
- Particle filter mask
- Respiratory face-masks, claiming to pertain to categories N95, KN95, and FFP2.
- Packaging description
- The product is packed in a plastic foil and sold in sets of 4 in a black cardboard box (w x l x d = 15 cm x 16 cm x 2.3 cm).
- bd BADO
- Starkit (auf der Verpackung), bd BADO (auf der Maske)
- Type / number of model
- Art. Nr. Mask N98-4
- Batch number
- Is the product counterfeit?
Withdrawal of the product from the market
Date of entry into force 01/02/2021
Year - Week
Country of origin