- Risk type
- Health risk / other
- Notifying country
- Alert number
The particle/filter retention of the material is insufficient (measured values: as low as 32%). The total filtration capacity of the mask is insufficient (measured mean values: as low as 65%). Consequently, the product does not fulfil the health and safety requirements; thus, even if combined with other recommended measures, it may not properly protect.
The product does not comply with the Personal Protective Equipment (PPE) Regulation and with the relevant European standard EN 149.
- Protective equipment
- Particle filter mask
- White protective respiratory filtration half-face mask, with elastic ear bands and nose strip, marked with "KN95".
- Packaging description
- The product is sold in packs of 5 items, packed in transparent plastic pouch with green trim measuring 200x130mm.
- KN95 Stereo Protective Mask
- Type / number of model
- Is the product counterfeit?
Recall of the product from end users
Date of entry into force 04/05/2021
Year - Week
Country of origin